LifeScan Announces Voluntary Recall of All OneTouch Verio IQ Blood Glucose Meters
Milpitas, CA – March 25, 2013 – LifeScan, Inc. is initiating a voluntary recall and replacement for all of its OneTouch VerioIQ blood glucose meters in the United States, effective immediately.
LifeScan is recalling and replacing all OneTouch VerioIQ Meters because at extremely high blood glucose levels of 1024 mg/dL and above, the meter will not provide a warning that the blood glucose is extremely high and will shut off, thereby potentially leading to incorrect treatment and delaying proper treatment.
The likelihood of experiencing an extremely high blood glucose level of 1024 mg/dL or higher is remote; however, when such a blood glucose level occurs, it is a serious health risk requiring immediate medical attention. Because these products do not provide an appropriate warning at glucose levels of 1024 mg/dL or higher, diagnosis and treatment of extreme hyperglycemia may be delayed or incorrect treatment may be given resulting in potentially serious health risk or fatality.